Language English. Keithley: DMM Instruction Manual. Identifier keithley_DMM__Instruction_Manual. Identifier-ark ark://t8kd5tt7c. Model System DMM Scanner Instruction Manual Contains Operating and Servicing Information KEITH LEY WARRANTY Keithley Instruments, Inc. warrants . Keithley Pdf User Manuals. View online or download Keithley Instruction Manual.
|Published (Last):||18 September 2013|
|PDF File Size:||2.7 Mb|
|ePub File Size:||11.83 Mb|
|Price:||Free* [*Free Regsitration Required]|
The following procedure explains how to use the zero feature to establish a reference: DC Voltage Measurements 2. IEEE address is programmable from the front panel.
Tektronix / Keithley – In Stock, We Buy Sell Repair, Price Quote
Note that the two outputs must not be conrvected together in the 4-pole mode. This com- mand will clear aU bits in the SRQ mask.
The allowable range for the trigger delay period is between 0 and In T6, the unit provides its own trigger. Readings will continue to update the display while the filter is settling, but the display will not represent the final, filtered reading value until FLTR stops flashing when the proper number of readings have been averaged.
If not, check to see that proper bus con- nections are made, and that the instrument is programm- ed for the correct primary address Place the instrument in “one-shot on external trigger” or “continuous on external trigger” mode as explained in paragraph 2. To disable data store and return to normal operation, press any function key. For complex, nonsinusodial waveforms, however, measurements made with an averaging type meter can be grossly inaccurate.
To send GTL, the controller must perform the following sequence: For example, if you take a lOV channel 1 reading on the 30V reading, the ratio reference will remain lOV on the V range.
Use uprange or downrange to select the desired inter- val keithhley, select or default, then press NEXT. Test front panel LEOs and annunciators. CAUTION Precision calibration signals must be connected to the instrument before attempting calibration, otherwise instrument accuracy will be affected.
L Select the range and function as required. Plug the keithle cord into the rear panel power jack and plug the other end of the cord into an appropriate, grounded power source.
Keithley: DMM 199 Instruction Manual
The unit will scan all channels in the set and then stop. The allowable range for selected in- terval is 15msec to Use uprange or downrange to select the desired interval operation, then press NEXT.
The pro- gram wiU set the store size to line 30enable the data store line 40turn on the data store output line 50and then request and display all lOO readings lines Wait for SRQ error. Factory Default Conditions Full text of ” Keithley: If a bus error occurs. Table shows the current output for each resistance range. The maximum ratio is 9. The controller sets ATN true. The location will wrap around to the opposite kwithley of the data store buffer once keithle lowest or highest location is accessed.
If an additional instruction manual is required, order the manual package Keithley Part Number Reading and buffer location with prefix. The OdB point is now established. Filter P Rate S Refer to paragraph 2.
Lists typical times when accessing instrument data over the bus. Instead of the more difficult method of adjusting a number of potentiometers, the user need on- ly apply an appropriate calibration sigrial and send the calibration value over the bus. For example, with 0. Some of the statements have two forms, with the exact con- figuration depending on the command to be sent over the bus.
Size The size of the data store can be controlled by one of the following I commands. F15X— Invalid command option because 15 is not an option of the F command.